I try the configuration of the impedance analyzer with the LCR meter extension board. The results for 60pF capacitor measurement (step : 100 , min frequency : 10KHz, max frequency : 6MHz ) is attached in the creen picture
is there someone to tell as what hapens between 1 and 2 MHz? Why the curve falls from 57pF to 38pF?
This experience was reconduced and I have the same result.
I try to measure the capacitor with a HIOKI 3632 LCR meter and this curve falls is the result of the Redpitaya measurement only.
A second question : there is like an over estimation of the capacitor and the phases. How can we calibrate the device ? I think that a phase can't be over -90° (négative résistor).
Placement, modules, components and accessories; the ones that exist and the the nice-to-be's
1 post • Page 1 of 1
jadalnie klasyczne ekskluzywne meble wypoczynkowe do salonu ekskluzywne meble tapicerowane ekskluzywne meble do sypialni ekskluzywne meble włoskie
Who is online
Users browsing this forum: No registered users and 4 guests