I use labVIEW and STEMlab 125-14 to collect the signal of 3V by selecting HV GAIN option and the jumper of board is set to HV. When the signal is 2V and I set the trigger level to 2.5V, the trigger should not work, but the return value of wait or trigger.vi is the flash of TD or error and the board is the status of trigger. When the signal and the trigger level is the range of +/-1V, it is normal. Why cannot the trigger level exceed the +/-1V selecting HV GAIN? Is this the bug of the firmware? Please give me some advice, thank you!
Placement, modules, components and accessories; the ones that exist and the the nice-to-be's
1 post • Page 1 of 1
- Posts: 1
- Joined: Mon Dec 06, 2021 4:55 am
jadalnie klasyczne ekskluzywne meble wypoczynkowe do salonu ekskluzywne meble tapicerowane ekskluzywne meble do sypialni ekskluzywne meble włoskie
Who is online
Users browsing this forum: Google [Bot] and 2 guests